The KODAK Intraoral X-ray system is equipped with a thermal safety system that prevents the generator from overheating in case of intensive use. Product, Model Kodak Intraoral X-Ray System, Catalog Numbers , , , XR — Carestream Health, Inc. Made in France Kodak Intraoral X-Ray System Performance. Operational ease. Safety. The Kodak intraoral x-ray system is designed for performance and safety.
|Published (Last):||1 December 2011|
|PDF File Size:||8.96 Mb|
|ePub File Size:||18.33 Mb|
|Price:||Free* [*Free Regsitration Required]|
There are two grooves on the plastic cup that must match those in the bottom of the housing. Interested in the regional page in your own language?
A short beep sequence is emitted at the end if the self-calibrations succeed. Dental Tribune Weekly Digest. Start by measuring the 13 Kohm feedback resistor located inside the housing intaoral Ohmmeter between ground and yellow small wire on the tubehead connector once removed from the arm. Optimized for both film and digital radiography usage, the system includes a redesigned electronic timer that makes the setting of exposure parameters failsafe.
Used & Refurbished Dental Equipment
Can be located anywhere from close to the tube to outside of the operating room Timer can also be mounted on the wall-framework. Log in or Register to rate and post a comment. Please Mr Faustino can you intraaoral me the service manual and im interessed to inyour spare parts can you mail me suissedental gmail.
Then insert the assembly gently inside the metal housing, taking care with the position. Content you may be interested in: Would be thankful if you could send me the tech manual as well.
During the Thomas P. Faustinio I x-rwy also having KV error and the system beeping. Live Webinar 19 Mar December 3, Live Webinar: On the power board set the dipswitch 4 of SW2 to ON Switch “ON” the board while maintaining the X-ray switch of the control panel activated until the end of the self-calibration sequence.
Is it possible if I can use a regular square ceramic resistor 5 watt? I have the same problem Can anybody send me the service manual for please my email address: Send me a mail and I will send you a link to view my technical files or give you more information.
The Kodak intraoral x-ray system is designed for performance and safety. If you wish to download it, please recommend it to your friends in any social system. And with multiple installation configurations available, you can easily adapt the unit to match the needs of your practice. To make this website work, we log user data and share it with processors. Designed for dental professionals seeking ease of use and reliability, it provides high quality results for accurate and safe diagnoses.
Afora is the first company to join the newly launched quip Labs, a venture studio created to foster innovation and development for emerging oral health products, platforms and services.
You mention that person needs a K control panel? Hello, You will receive a link from Hightail to see my K files, Regards.
Kodak Kv/error – Service Technicians Forum
About project SlidePlayer Terms of Service. It gives KV Error along with bip and on the control panel two green lights turn on. The latest KODAK intraoral dental X-ray system is designed to deliver new ease of use and dose management capabilities.
I thinks it will fit under the tube board but not sure the effectiveness of it. Just check if it’s ok 13 Kohm and MOhn and solder unplagged wire to it. For next procedure for self-calibration: Hi, KV error on a means either a power board failure or a tubehead failure which is the most frequent problem. Additionally, the dose level is displayed after each exposure so radiation dose can be tracked and monitored.
KODAK 2200 intraoral X-ray system is unveiled
Changes in ABO 22200 Unsolder the wires connected to internal socket and lift up the tube board carefully. Hi Faustino, Do you have ceramic resistor 13 Kohm? Chad Gehani, an endodontist, as its new leader. Sign In or Register. Related Searches Flat panel detector Dental imaging software Medical radiology software Cephalometric analysis software Dental software Simulation software Analysis software.
I forgott to send you my email: You need the K control panel to service the unit. Heating the assembly with hair dryer helps sometimes. Once the old solder removed, the wires soldered back with new solder do not overheat place the high voltage kpdak spring and insert the high voltage multiplier block inside the plastic cup containing the feedback resistor. Limited Ducted Split Systems